Get Empowered: Tip #239, Empower Walk-Up Interface
Tip #239: How to Use the Walk-Up Interface in Empower
Welcome back to Get Empowered! In the last Empower tip-of-the-week blog for Empower Chromatography Data System, we revisited our VLOG series showcasing Empower as an important tool to implement Method Lifecycle Management (MLCM) and learned about a tool to help monitor method performance called Empower Data Science Link, powered by TetraScience (Episode #13 /Tip #238).
In this week’s tip, we will begin a new blog series on how to use the Walk-Up Interface in Empower.
The Walk-Up Interface is a simplified interface in Empower for easy acquisition and processing of samples. It is well suited for high thru put environments. All the configuration and method creation are done prior to the user accessing the system to keep things simple.
The user in this case only has access to the Walk-Up Interface.
Let’s get started.
In this first example, we want to run one sample, and have it quantitated against a calibration curve (this could be LC, GC or GPC).
Double click the Empower shortcut, Enter ‘User Name’ and ‘Password’, click ‘OK’ (figure 1).
Select a ‘Project’ on the left and select a ‘chromatographic system’ on the right, click ‘OK’ (figure 2).
The Walk-Up Interface opens and defaults to ‘Run Samples Only’ (figure 3).
Enter a ‘Sample Name’ and select a ‘Method Set’ (NOTE: The run mode is determined by which methods are in the Method Set. For example, if you want to Run and Report, you must select a Method Set with an Instrument Method, Processing Method and Report Method). Enter an ‘Injection Volume’, enter a ‘Run Time’, and a ‘Sequence Name’. The ‘Options’ button allows entry for ‘Dilution’ and ‘Sample Weight’, if needed. Click ‘OK’ and click ‘Run Samples’ (figure 4).
If the MS Mass Editor window appears, click ‘OK’ (figure 5).
If you are in ‘Run and Report’ mode, select the ‘printer’, click ‘OK’ (figure 6).
A message appears as a reminder to add the vial in the autosampler. Click ‘OK’ (figure 7).
In the lower left corner of the screen, you will see the message change from ‘Setting Up’ to ‘Waiting for Injection’ (figure 8).
During the run, the message displays ‘Injection Running’ and eventually will display ‘Complete’ once the run is finished (figure 9).
In the second example, we want to run a series of samples using a Sample Set Method Template.
Once logged in using the Walk-Up Interface, select ‘Run Samples Using a Template’. Highlight the template from the list, enter a ‘Sequence Name’ and click ‘Run Samples’. The Variable window opens prompting to enter the necessary information, in this example, it is the ‘Number of sample vials’ and ‘Vial starting position in tray’. Click ‘Next’ (figure 10).
Continue entering any necessary information and on the last screen, click ‘Finish’. The sequence will begin (figure 11).
It’s that easy!
Please rate this Empower Tip of the Week
Next week in Empower Tip #240: We will continue our series on using the Walk-Up Interface.
JOIN US for an Empower: Ask Me Anything Webinar on September 22, 2021
Neil will answer your questions about streamlining your Peak Purity workflow with Empower using the PDA and QDa Mass Detector.
More Empower Resources:
- Get the latest Empower projects, options, and more. Check out the New Waters Marketplace.
- Want to learn more? Check out our Empower training courses and eLearning.
- Minimize your lab downtime and protect your business against potentially expensive disruptions with the Empower Enterprise Software Extended Critical Support Plan with 24 x 7 x 365 coverage – Learn more
- Build your technical skills and professional network. Share, ask, and interact with peers from around the globe in dynamic discussions by joining our Informatics Community Forum.
Do you want Empower Tips sent to your inbox every week?
Questions? Tips of your own? Let us know!