Tip #49: How to subtract a baseline from a 2D chromatogram by Deriving a Channel (part 3) in Empower Software.
Tip #48: Learn how to correct for the time offset between detectors that are connected in series by Deriving a Channel in Empower Software. (Part 2)
Tip #47: Smoothing a channel of chromatographic data collected with a UV detector through Empower Software’s Deriving Channels feature.
Tip #46: Recently I’ve had several questions about how baselines are determined using Empower Software’s ApexTrack peak integration algorithm. Let’s take a look at how they work.