Wouldn’t it be nice if Empower Software could flag a failing result from a chromatography run? It’s actually easy to set up. See how in tip 27.
This tip will address how to perform cross-channel internal standard calibration and quantitation.
This tip will address how to perform internal standard calibration and quantitation using Empower Software.
This tip for using Empower Software addresses how to add chemical structures to peaks in either Review or the Report Method.
This tip will address how to add Timed Events interactively in the chromatogram while developing a Processing Method in Empower Software.
This tip will address how to set the peak width and detection threshold when working with the ApexTrack peak detection/integration algorithm.
This tip will address how to set the Peak Width and Threshold when working with the Traditional peak detection/integration algorithm.
In this tip for using Empower Software, we address comparing UV spectra from different channels.
Wouldn’t it be nice if when you bring 3D PDA data into Review it automatically displayed a chromatogram? And the chromatogram was integrated with lambda max for each peak displayed? Guess what? Empower can be set up to do this and it is easy.
Empower Software Tip #18: Simplifying Review when Working with 2D LC Data