In Empower Software, a Peak Custom Field is a user-defined calculation pertaining to a peak or peaks within the chromatogram. Our goal in this Empower Tip is to calculate the percent of Label Claim for an ingredient in the final product.
As we continue our series on using Custom Fields in Empower Software, learn how to create a Component Custom Field.
With Empower Tip #83, we continue looking at Custom Fields and learn how to create a Custom Sample Set Field. A Sample Field is something that identifies a sample and each sample can have its own set of unique identifiers. A Custom Sample Set Field is something that identifies the entire Sample Set. Common Custom Sample Set Fields are method type, column used, and mobile phase used.
Empower Software’s Result Audit Viewer makes it easier to review the audit trail information associated with a Result Set by bringing this information into one window rather than going to all of the individual audit trail windows. In this tip, we take a tour of the Result Audit Viewer.
Lots of questions about calculating impurities in Empower Software! Here’s another answer to a reader who inquired: How do you quantitate impurities against the main component? We’re happy to show how…
#69: Learn how Empower Software calculates Peak-to-Peak Noise for USP, EP, and JP Signal-to-Noise calculations.
We begin a multi-part discussion about the System Suitability option in Empower Software, starting with what are Statistical Moments and how are they helpful.
#67: How to save extracted channels after 3D data processing in Empower Software.
#66 This week we will begin a new multi-part discussion around using Method Sets in Empower Software. This post shows how to subtract a 3D baseline from 3D PDA data.
#65: In this final tip in our Deriving Channels series for using Empower Software, we learn about Switch To, a function that you can use to combine data from channels generated by different wavelengths or detectors into one, so that you process a single chromatogram.