Why it is important to align chromatograms when collecting data from both a PDA and a QDa mass detector?
Let me show you.
How do I extract chromatograms and spectra while in the Mass Analysis window?
I thought you’d never ask. Learn how.
Did you know that you could work with MS data from an ACQUITY QDa Mass Detector in Empower? Read more and learn how.
Need to know when to batch process the sample set to get results from Custom Fields in Empower? Read more to learn how.
Want to know how to use the average area of replicate injections of a sample to quantitate my analyte(s) of interest? Learn how to do this using Empower Software.
Learn how to report the maximum pressure during a chromatographic run when using Empower Software.
How can you report the pressure or temperature at a certain time point within the chromatogram? See how to use a Custom Field in Empower Software to do this.
What’s the difference between Summarize Custom Fields and Summarize Custom Fields Incrementally in Empower Software?
This week, we will create an Intersample Custom Field with a summary function, such as average or %RSD, and learn how Empower Software calculates the average area or the %RSD of the area (i.e., for a peak of interest over multiple samples).
What if we wanted to make a calculation in Empower Software involving a peak in one chromatogram and a peak in another chromatogram? In this case, you’d use an Intersample Custom Field. See how to apply it.